The unit is made by Fine Offset Electronics. Based on information from Richard Barsby it is probably the WH1080 as he owns a WH1081 which he says doesn’t have the DCF time sync and to me at least the two are visibly indistinguishable on the Fine Offset Electronics website. I’ve also seen what appears to be the same unit sold as Elecsa AstroTouch 6975 by a German seller and Zephyr PWS-1000TD from the USA on eBay. PCE-FWS 20 is another incarnation, It is also being sold by Maplin as a USB Wireless Weather Forecaster for £79.99 which made it one of the most common entry level weather stations in the UK.
G6KIZ Weather station
I’d been considering adding weather data to the G6KIZ iGate for some time but really couldn’t justify the cost. Then M0XDK drew my attention to the Watson W8681 weather station which W&S had announced in their newsletter and were selling on eBay at £95.95 delivered. The features on offer would normally cost at least 3 times more. Obviously there would be some corner cutting but I decided to get one anyway – an interesting gadget regardless of whether or not the data could be pushed into UIView. It arrived by courier next day which was excellent.
The W-8681 sensors
The sensors are intended to be assembled on the short two section stub mast that is included. This is very easy. Two of the leads are much longer than necessary for this layout and I used zip ties to make as tidy a job of the wiring as possible. Those long cables came in handy as I later modified this mounting in light of experience by attaching the wind gauges to a longer pole (actually a metal broom handle), ensuring that the transmitter which houses the temperature gauge is in the shade, and mounting the rain gauge to something firm so that it can’t be buffeted by the wind leading to false rainfall indications. A picture of this layout is shown in my WH1090 review – a later Fine Offset weather station.
I have some concerns on just how weatherproof the push in modular connectors and battery bay will be. The specifications say waterproof to IPX3 which may or may not prove to be good enough.
Two AA batteries need to be fitted and these should last at least two years according to the specifications.
Once the sensors were assembled on the stub mast I just zip tied the mast to an 8 foot length of rough cut timber and screwed it to a fence post. The mounting position is a bit of a compromise as it is probably a bit too close to a large conifer for wind direction measurements to be spot on but I doubt that the unit would ever be especially accurate anyway as the vane is very lightweight and completely undamped so the slightest gust sets it swinging.
The W-8681 display unit
Attention now shifted to the quite large display. It transpired that what I thought was a loose protective covering was in fact a faulty membrane that covers the screen to provide the touch screen facility. The unit was returned to W&S for replacement which took over a week to arrive which was far from good and rather took the gloss off the original fast delivery.
It requires three AA batteries which are supposed to last a year. I found that the contrast and backlighting (green LEDs) were both rather poor. M0XDK’s had a flickering led – caused by poor quality soldering and OK now he has fixed it. This unit is probably where the inevitable cost cutting is most immediately obvious. Later units incorporate a contrast adjustment and one correspondent says that his is fine on the middle setting.
This operates at 868MHz and I was originally concerned that this may be a problem as DigitalHam Towers is an old stone building with walls over two feet thick. It seems so far that I need not have worried as with the new display it has maintained sync.
The unit also uses the DCF transmitter located near Frankfurt in Germany to control its clock display. This took an age to sync initially and when it did it showed an hour fast despite the time zone being set correctly. I can only assume that this is a bug and it has added an hour DST despite it being December. I adjusted this manually.
The supplied software automatically adds to an ever growing file with each record being a comma separated list of values. To get this data into a format acceptable to UIView requires some detective work on what is what and scripting to reformat it. As it happens I have a text editor which can run REXX which I used for this purpose. I doubted that this was mainstream enough to be of any use to other hams so I have now produced a Python script which will create a UIView data file – EasyWeather to UIView script This isn’t exactly mainstream either but at least it is open source.
The EasyWeather software has some problems with long term running. I have seen it hang once with an out of resources error. It regularly loses parts of the window furniture and wind direction display. Simply closing it and restarting clears these problems up but it does take a little while to get started again – this delay will probably increase as the internal data buffer fills because the documentation mentions that reading this data from the unit can take a couple of minutes. Changes to the various settings only seem to affect the display and the output data is fixed units – something to be aware of when converting for use in UIView.
W8681 User Guide for anyone who has lost or did not receive a copy.
So is it worth the money? I’d give that a reserved yes. That may turn to a definite yes providing water ingress or other weather damage doesn’t put paid to the sensors. If I hadn’t been able to use the data for UIView and was judging the unit just on the display and software I would find it hard to recommend. If your main aim is use as an online weather station and you cannot do the necessary coding you would be advised to avoid this unit until someone has done the work and posted it on the web. I am rather surprised that Waters and Stanton have not produced software for this purpose as it would surely make the unit much more attractive to radio amateurs.
To see what happened later read the follow up.
I’ve had lots of correspondence about this weather station so now that I’ve converted DigitalHam to use WordPress I’m posting some of the emails as comments.
Bravery is being the only one who knows you’re afraid.